Yozil All goods are sold as seen and will need thoroughly hamge before use. Please check your email account for more details. Unfortunately, your registration has been declined by the auctioneer. The Buyer must provide to the Seller satisfactory proof of shipment within 14 days of removal otherwise the Buyer shall be liable for the outstanding amount of VAT.
|Published (Last):||9 August 2010|
|PDF File Size:||5.11 Mb|
|ePub File Size:||15.82 Mb|
|Price:||Free* [*Free Regsitration Required]|
Hi there, I was lucky enough to be given an old Hameg HM oscilloscope by my college. I am very competent when it comes to soldering though, so swapping parts out is no problem. The scope switches on fine, but to get the screen to show a trace, I have to VERY carefully and slowly tune the Y-POS knob to a specific, tiny range, at which point the line quickly comes up from the the bottom of the screen. As the trace approaches the centre of the screen, it gets wobbly and unstable before eventually shooting off.
If I can persuade the trace to come onto the screen, adjusting the FOCUS knob will move the trace up and down as well as adjust the focus as expected although it disappears off the screen before I can see a significant difference in focussing. The usable range of Y-POS increases and I can get the trace centred and move it trace up and down more easily.
If I move it more than a division away from centre though, it becomes curved and then disappears. This behaviour is identical on both channels. Would this be a good place to start? Thanks in advance for any help -d Last edited by d0pefish; at PM..
Second-hand Hameg for sale on UK's largest auction and classifieds sites
Hameg HM 203 Manual
Download Hameg HM203-7 docs - Various electronics service manuals
HAMEG Analog Oscilloscope 20mhz Hm203-7 H28ded62