Samurisar Image not available Photos not available for this variation. We always declare the actual purchase value as declaring lower purchase value is illegal by international law. Sell now — Have one to sell? This amount is subject hanuoung change until you make payment. Minimum monthly payments are required.
|Genre:||Health and Food|
|Published (Last):||4 March 2013|
|PDF File Size:||16.84 Mb|
|ePub File Size:||7.90 Mb|
|Price:||Free* [*Free Regsitration Required]|
Please check whether the product is exactly the same as you ordered. Before using the product, please read this instruction manual carefully. Please keep this manual where you can view at any time. Safety information Alerts declared in the manual are classified to Danger, Warning and Caution by their criticality.
Danger Do not touch or connect any undesirable conductive part to input-output terminal, since there is a possibility of electric shock. Malfunction or an incorrect operation may be a cause of leading to a serious accident.
Fuse rating : V 0. There is a possibility of malfunction, electric shock or a risk of fire. This is a cause of electric shock, malfunction, or failure. H without condensation. There will be a temperature error if a general lead is used.
There will be a temperature error if the 3 lead wires do not have the same resistance. If it is not possible, please use shielded wires for the input signal wires. There is a possibility of malfunction of the product by electric leakage, if a grounded sensor is used. The noise filter should be grounded on the panel and the wire between the output of the noise filter and the power of the instrument should be as short as possible.
In this case, SSR output is recommended. Please use a delay relay together when it is used as a signal on the outside of interlock circuit or others.
Hanyoung Temperature Controllers
Hanyoung HX4 Instruction Manual
HANYOUNG NX9 PDF
Hanyoung Nux Co., Ltd